|Measuring range||0.32 to 32,000 mg/L S2-|
(10-5 to 1 mol/L) S2-
|pH range||3.2 mg/L (10-4 mol/L) S2-|
pH 12 to 14
|Applicable temperature range||0 to 50℃|
|Response time (90%)||Within 10 seconds|
|Applicable reference electrode||2060A,|
|Selection coefficient||CN- = Not acceptable|
S2O32- = 10
l-、F-、Cl-、PO42-、SO42- = 1,000
• The response time is the time which is required to reach 90% response when the ion concentration is gradually changed from 10-4 mol/L to 10-2 mol/L with the solution stirred.
*1: 90% response when ion concentration is changed to 10-6 mol/L ∼ 10-2 mol/L
*2: 95% response when ion concentration is changed to 10-3 mol/L ∼ 10-1 mol/L
*3: 95% response when ion concentration is changed to 10-4 mol/L ∼ 10-2 mol/L
*4: 95% response when ion concentration is changed to 10-4 mol/L ∼ 10-1 mol/L
• The selection coefficient is a ratio of the limit concentration of coexisting ions (mol/L) to the ion concentration to be measured (mol/L); The value of 1000 means that the coexisting ions can be permitted up to 1000 times the ion measured and ''not acceptable'' means that chemical change occurs in the solid response membrane.
Do you have any questions or requests? Use this form to contact our specialists. * These fields are mandatory.