Water Distribution Monitor
|Measurement items||(Standard configuration) turbidity, color, free residual chlorine, water pressure, pH|
(Optional items) conductivity, water temperature
See Table 1.
See Table 1.
See Table 1.
See Table 1.
Automatic zero calibration
|(Turbidity, color, free residual chlorine)|
Calibration method: Filtrate sample water
Calibration start: Interior…started by internal timer
Exterior...started by external contact input
Calibration cycle: 5 hours to 9999 hours (user setting)
Calibration time: Approx. 15 minutes
|Automatic draining||Draining start: Interior...started by internal timer|
Draining interval: 5 to 9999 minutes (user setting)
Sample water condition
|Temperature: 0°C to 40°C (do not freeze)|
Pressure: 0.1 MPa to 0.75 MPa
Conductivity: 10 mS/m or higher
Analyzing unit input rate (flow rate): 50 mL/mi to 100 mL/min
● In test operation, flush the meter thoroughly before running water.
● Make sure to set up a bypass for piping to the meter.
● If sample water may freeze, take measures to insulate the unit from cold and to retain heat.
● Sample Water measured with this device cannot be distributed.
|Ambient temperature: 0°C to 40°C|
Ambient humidity: 85% or lower
|Type: Measurement values: Number of outputs corresponds to number of measurement values (up to seven values including five values of the standard configuration)|
Specifications: 4 mA to 20 mA DC, insulated output (non-insulated between items)
Maximum load resistance: 600 Ω
|Contact output||Types: Power failure, batch alarm, maintenance|
Contents: Power failure…occurs when the power is turned off
Batch alarm…cell temperature adjustment abnormality, temperature compensation abnormality,
cell wiper abnormality internal communication abnormality, leak, battery abnormality,
concentration upper limit concentration lower limit,concentration device upper limit,
concentration device lower limit, light source abnormality zero calibration, span calibration, analyzer abnormality
Maintenance…when the system enters maintenance or calibration mode
Specifications: No-voltage contact output, a contact interface
Contact rating: 125 V AC 0.3 A, 30 V DC 1 A (at resistance load)
Each output is an independent COM interface.
|Type: Cleaning start, zero calibration start, alarm check setting, abnormal water sampling|
Contents: Cleaning start…started by closed contact input
Zero calibration start…started by closed contact input
Alarm check setting…started by closed contact input
Abnormal water sampling... started by closed contact input
Specifications: No-voltage contact input (open collector connection is possible), insulated input
ON resistance: maximum 100 Ω
Open voltage: 24 V DC
Short-circuit current: maximum 13 mA
|Interface: RS-232C compatible|
Communication speed: 19200 bps
|Integrating function for flow rate used in the system (counting type)|
Internal leak detecting function
|Measurement data is stored on system, and can be transferred to a CompactFlash®* (CF Card).|
Data saving interval: 1 minute or 1 hour
Data saving frequency: 1 minute: Saves at every 0 second of the minute
1 hour: Saves at every 0 second of the hour
Data memory time: 1 minute interval…for approx. 10 days
1 hour interval…for approx. 1 year
The latest data will be stored.
|Wiring connector||Wiring inlet|
Compliant cable: 12.5 mm dia. to 14.5 mm dia.
Power source: 4.5 mm to 6 mm
|Pipe connector||Sample inlet: Rc1/4|
Condensate outlet: Rc1/8
Condensate outlet (internal): 6 mm dia. hose nipple
Air inlet: Rc1/8
Condensate outlet (for detection): 6 mm dia. hose nipple (Rc1/8)
Calibration inlet: Rc1/8
|Installation||Designed for indoor installation.|
* For outdoor installation, a case is required (option).
|Power supply||90 V to 264 V AC: 50/60 Hz|
|100 V to 240 V AC: Maximum 120 VA|
Approx. 18 kg
350 (W) x 160 (D) x 420 (H) (unit: mm)
Munsell 5PB 8/1
● Flat and stable location with minimum vibrations or shocks
* CompactFlash® is a registered trademark or trademark of SanDisc Corporation in the United States and other countries.
|Measurement items||Measurement method||Measurement range||Repeatability||Calibration method|
|Turbidity||90 degree light scattering method||0 to 2, 0 to 5, 0 to 10 NTU||±2.5% of full scale||Formazin standard solution|
|Color||Transmitted light absorption method||0 to 10, 0 to 20 TCU||±5.0% of full scale||Standard color solution|
|Free residual chlorine||Polarographic method||0 to 5 mg/L||±2.5% of full scale||DPD colorimetric method|
|Water pressure||Semiconductor detection method||0 to 1 MPa, 0 to 10 Bar||±1.0% of full scale||Standard pressure gauge|
|pH||Glass electrode method||2 to 12 pH||±0.1 pH||pH standard solution|
■Option (each parameter can be added to the standard type)
|Measurement items||Measurement method||Measurement Range||Repeatability||Calibration method|
|Conductivity||AC 2 pole method||0 to 500, 0 to 1000 μS/cm||±2.0% of full scale||KCI standard solution|
|Water temperature||Thermistor method||O to 50°C||±0.5°C||Standard thermometer|