|Product name||Converter for peracetic acid monitor※1|
|Temperature compensation element||Platinum resistance thermometer (Pt1000) 1000 W at 0℃, 3850 ppm/℃|
|Measurement range||100 to 4000ppm (0.010 to 0.400％)|
|Transmission output||Number of outputs: 2 (Negative terminals of each transmission output channel are connected inside and thus have the same electric potential.) 4mA to 20mA DC or 0mA to 20mA DC input/output isolated type Load resistance: 900 (max.) Linearity: Within 0.08 mA (Output only) Repeatability: Within 0.02 mA (Output only) Output setting: Peracetic acid concentration (Selectable from 0 to 22000 ppm or 0.000 to 2.200) Temperature (Selectable from 10C to 110C) HOLD function: Selectable from the previous value hold and the optional value hold (Setable without HOLD only while calibration|
|Contact output||Number of outputs: 5 Contact output: 4 (R1 to R4) Contact type: Non-voltage contact output, relay contact SPST (1a) Contact capacity: 240 V AC 1 A, 30 V DC 1 A (Resistance load) Contact function: Selectable from Upper/lower limit alarm (ON/OFF control) at each measuring objects(concentration, temperature) (Alarm Setting of delay time about measuring objects: 0 seconds to 600 seconds) Contact action: closed when status is in the event. (However, R1 to R3 share a common terminal. R4 and RF share a common terminal) Error alarm contact output (RF): 1 Contact type: Non-voltage contact output, Relay contact, SPDT (1c) Contact capacity: 240 V AC 1 A, 30 V DC 1 A (Resistance load) Contact function: Fail Contact action: Closed when status is normal for C-NO output. Opened when any erroneous status is detected or power is down for C-NO output. C-NC output is opposite action of C-NO. R4 and RF share a common terminal.|
|Contact input||Number of input: 1 (external HOLD input) Contact type: No-voltage a contact for open collector Condition: ON resistance: 100 ・ (max.)|
・Open-circuit voltage: 24 V DC
・Short-circuit current: 12 mA DC (max.)
|Communication output||RS-485 input/output|
|Self-diagnosis function||Calibration error /Temperture sensor error/Meter error|
|Power supply||24 V DC ±10%|
|Structure||Indoor-use panel installation type Panel case: ABS, Terminal: PBT Panel: IP65 dust and water proof structure|
|Conforming standards||CE Marking EMC Directive: EN61326-1※2 Class A, Industrial electromagnetic environment FCC rule FCC Part15|
|Product name||Peracetic acid sensor※3|
|Measurement principle||Electrochemical measurement system|
|Measurement target||Peracetic acid (CH3COOOH) ※4|
|Sample temperature range||0℃to 45℃ (without freezing)|
|Supply flow rate||500 mL/min to 2000 mL/min※5|
|Pressure||Within 0.1 MPa (constant)※6|
|Responsibility||(To the measurement cell after a supply start) T90 < 5 min※5|
|Ambient temperature||0℃ to 45|
|Liquid contact material||PVC, PTFE, Silicon, PPO, OPP|
|Internal liquid||Neutral phosphate(Food additive)|
|Cable length||5 m to 50 m max.※8|
※1 This product does not use for medical equipment.
※2 When the sensor cable, the transmission cable, or the contact input cable is extended to 30 m or longer, the surge test specified in the EMC directive for CE marking is not applied.
※3 This product does not use for medical equipment.
※4 Measurements may be affected by a change of the concentrations of hydrogen peroxide
※5 When supply flow rate decreases, responsibility may become slow. Supply flow rate is responsibility when switches from the solution including peracetic acid from the solution which does not include peracetic acid.
※6 When sample pressure includes a change, influence is reflected on instructions level. Install it to become the constant pressure.
※7 Under the constant ambient temperature, sample temperature and flow rate
※8 When the connection cable is extended to 30 m or longer, the surge test specified in the EMC directive for CE marking is not applied.